Payment & Shipping Terms:
|Method:||Gold Colloidal||Format:||5 Multiple Panel|
|Specimen:||Urine||Testing Time:||5-15 Minutes|
|Shelf Life:||24 Months||Application:||5 Drugs|
self testing drug kits,
accurate drug test kits
MultiDrug of Abuse Rapid test,Urine Specimen, High Accuracy, Easily & Quick operation on spot, Gold colloidal method
The Multi-Drug Rapid Test Panel with Adulteration is a rapid chromatographic immunoassay for the detection of drugs and drug metabolites in urine at the following cut-off concentrations in urine:
|Product||Cat No||Specimen||Sensitivity||Format||Kit Size|
|Drugs of Abuse (DOA) Tests|
|AMP||YD101S3||Urine||1000 ng/mL||3mm Strip||100T|
|BAR||YD201S3||Urine||300 ng/mL||3mm Strip||100T|
|BZO||YD301S3||Urine||300 ng/mL||3mm Strip||100T|
|COC||YD401S3||Urine||300 ng/mL||3mm Strip||100T|
|MOR(MOP)||YD501S3||Urine||300 ng/mL||3mm Strip||100T|
|MET||YD601SM||Urine||500 ng/mL||3mm Strip||100T|
|MTD||YD701S3||Urine||300 ng/mL||3mm Strip||100T|
|MDMA||YD801S3||Urine||500 ng/mL||3mm Strip||100T|
|THC||YD901S3||Urine||50 ng/mL||3mm Strip||100T|
|BUP||YD111S3||Urine||100 ng/mL||3mm Strip||100T|
|PCP||YD121S3||Urine||25 ng/mL||3mm Strip||100T|
|K2||YD131S3||Urine||200 ng/mL||3mm Strip||100T|
|OPI||YD141S3||Urine||2000 ng/mL||3mm Strip||100T|
|TCA||YD151S3||Urine||1000 ng/mL||3mm Strip||100T|
|KET||YD161S3||Urine||1000 ng/mL||3mm Strip||100T|
|TML||YD171S3||Urine||100 ng/mL||3mm Strip||100T|
|DOA 2-in-1||YD021C5||Urine||See Insert||2-in-1 Panel||25T|
|DOA 3-in-1||YD031C5||Urine||See Insert||3-in-1 Panel||25T|
|DOA 4-in-1||YD041C5||Urine||See Insert||4-in-1 Panel||25T|
|DOA 5-in-1||YD051C5||Urine||See Insert||5-in-1 Panel||25T|
|DOA 6-in-1||YD061C5||Urine||See Insert||6-in-1 Panel||25T|
|DOA 7-in-1||YD071C5||Urine||See Insert||7-in-1Panel||25T|
|DOA 8-in-1||YD081C5||Urine||See Insert||8-in-1Panel||25T|
|DOA 9-in-1||YD091C5||Urine||See Insert||9-in-1 Panel||25T|
|DOA 10-in-1||YD011C5||Urine||See Insert||10-in-1 Panel||25T|
DIRECTIONS FOR USE
Allow the test, urine specimen, and/or controls to reach room temperature
(15-30ºC) prior to testing.
Bring the pouch to room temperature before opening it. Remove the test panel from the sealed pouch and use it within one hour.
Remove the cap.
With the arrow pointing toward the urine specimen, immerse the test panel vertically in the urine specimen for at least 10 to 15 seconds. Immerse the dipstick to at least the level of the wavy lines, but not above the arrow on the test panel.
Replace the cap and place the test panel on a non-absorbent flat surface.
Start the timer and wait for the colored line(s) to appear.
The result should be read at 5 minutes. Results may be stable up to 1 hour after test initiation.
INTERPRETATION OF RESULTS
(Please refer to the illustration above)
NEGATIVE:* A colored line appears in the Control region (C) and colored lines appears in the Test region (T). This negative result means that the concentrations in the urine sample are below the designated cut-off levels for a particular drug tested.
*NOTE: The shade of the colored lines(s) in the Test region (T) may vary. The result should be considered negative whenever there is even a faint line.
POSITIVE: A colored line appears in the Control region (C) and NO line appears in the Test region (T). The positive result means that the drug concentration in the urine sample is greater than the designated cut-off for a specific drug.
INVALID: No line appears in the Control region (C). Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for Control line failure. Read the directions again and repeat the test with a new test card. If the result is still invalid, contact your manufacturer.
A procedural control is included in the test. A line appearing in the control region
(C) is considered an internal procedural control. It confirms sufficient specimen volume, adequate membrane wicking and correct procedural technique.
Control standards are not supplied with this kit. However, it is recommended that positive and negative controls be tested as good laboratory practice to confirm the test procedure and to verify proper test performance.
|ORIENT NEW LIFE MEDICAL CO., LTD.|
|Email:||Jerry @ newlifebiotest .com|
Contact Person: Jerry Meng