Payment & Shipping Terms:
|Properties:||Laparoscopic Ultrasonic System||Application:||Thoracic,Bariatric,Colorectal,Gynecologic,General Surgery|
|Compatible:||OR14 OR23 OR36 OR45||Cutting Diameter:||5mm|
|Suturing:||Suturing Thickness 2mm,1.5mm,1mm||Sterilization:||ETO Sterilization|
|Character:||Adaptive Technology||Tip:||Curved Tip|
eto sterilization ultrasonic harmonic scalpel,
23cm shaft length ultrasonic harmonic scalpel,
23cm shaft length ultrasonic scalpel
Laparoscopic Surgery Harmonic Ultrasonic Surgical Scalpel
The ultrasonic scalpel is a surgical instrument used to simultaneously cut and cauterize tissue. It is the latest device to have been introduced in last decade or so. Ultrasonic energy is famously used in Harmonic scalpel where ultrasonic energy is converted to mechanical energy at the active blade. The main mechanism is the active blade which delivers high-grade frictional force while the inactive upper arm holds tissue in apposition. Its main advantages include precise dissection, reliable hemostasis, less lateral thermal spread and charring mainly works by applying pressure and then sealing with a denatured protein coagulum while applying ultrasonic vibration to denature hydrogen bonds perform vessel coagulation.
Transfer and amplification ultrasonic mechanical energy from the transducer handles to the organization, to ensure that appropriate clamping force, implementation of check hemorrhagic cutting and/or the organization setting.
The instrument includes a manual switch adapter (MIN, MAX manual switch) and a wheel-integrated locking wrench.
The Ultrasonic Scalpel System is indicated for soft tissue incisions when bleeding control and minimal thermal injury are desired. The System instruments can be used as an adjunct to or substitute for electrosurgery, lasers, and steel scalpels.
|Description||Code||Shaft Diameter||Shaft Length||Tip|
|Harmonic Scalpel Shear||OR14||5mm||14cm||Curved|
Refer to a compatible Harmonic Generator User Manual for hand piece attachment and system operation instructions.
1. Connect the assembled Hand Piece and instrument to the generator and turn the generator power on. Do not turn the generator power on before the Hand Piece and instrument are connected to the generator.
2. Select the desired variable or minimum power level using the INCREASE and DECREASE buttons on the generator.
Minimum starting power level defaults to power level 3. For greater tissue cutting speed use a higher generator power level, and for greater coagulation use a lower generator power level. The amount of energy delivered to the tissue and resultant tissue effects are a function of many factors, including the power level selected, blade characteristics, grip force, tissue tension, tissue type, pathology, and surgical technique.
MAX power is set at power level 5 and cannot be adjusted.
3. The Ultrasonic Shears instrument may be operated with either the foot switch or hand control. For foot switch or hand control function, refer to a compatible Harmonic Generator User Manual for further detail and setup and operation instructions.
4. For optimal performance, clean the instrument blade and clamp arm throughout the procedure by activating the instrument tip in sterile saline. The instrument can be wiped with a sterile moist gauze sponge to remove tissue, if necessary.
WARNING: Do not touch the instrument to metal while activated. See Warnings and Precautions.
5. If tissue is still visible in the clamp arm, use hemostats to remove residue.
6. The blade is ultrasonically energized when either the foot switch pedal is depressed or one of the hand controls is depressed.
Pressing either the left foot pedal of the foot switch or the proximal hand control (MIN) on the instrument activates the selected minimum power level.
Pressing either the right foot pedal of the foot switch or distal hand control (MAX) on the instrument activates the maximum power level.
The generator provides an audible tone to indicate when the instrument blade is active.
The generator changes to a second activation tone as Adaptive Tissue Technology regulates the delivery of energy.
Thermal influences such as fluids or minimal to no tissue in the jaws may affect the presence or timing of the tone change.
The tone change does not provide confirmation of tissue effect. When the second tone is heard, the situation should be assessed and the intended surgical action completed, such as gradual application of tension to facilitate transection.
The secondary activation tone change is not a substitute for surgical experience.
WARNING: Avoid accidental contact with other instruments during use. Scratches on the blade may lead to premature blade failure. See Warnings and Precautions.
1. Q:Are you a factory or trading company?
A: We are a factory, we have rich experience for manufacturing Medical Device more than 10 years.
2.Q:Where is your factory located? How can i visit there?
A: Our factory is located in Jiaxing city, Zhejiang Province, China.
It is about more than 1 hours from Shanghai.
2. 3.Q:Can you make samples?
A: Yes, we can make samples for you.
4.Q:How does your factory do as quality control?
A: Quality is the lift of our factory. We have inspector in different post, and there are three inspectors before packing to get best quality products.
5.Q: How is your post market service ?
A: If any damaged rings in the course of carriage, with the checking it's our mistake, we will take full responsibility to compensate the customers.
6.Q:Can you do OEM for us?
A: Certainly. We also can build the moulding for our new customer.
7.Q:Does the price include freight?
A:No, please choose the express service provider or negotiate with us.
Contact Person: Jerry Meng