Payment & Shipping Terms:
|Category:||HOME Fertility Test||Format:||Cassette|
|Specimen:||Serum/urine||Testing Time:||5-15 Minutes|
|Shelf Life:||24 Months||Application:||HCG Pregnancy Test Home|
female fertility test kit,
reliable pregnancy test kit
High Accuracy HCG Pregnancy Serum Test Home Fertility Testing Kits, gold colloidal method urine/serum specimen
hCG One Step Pregnancy Test Cassette (Urine/Serum) is a rapid chromatographic immunoassay for the qualitative detection of human chorionic gonadotropin (hCG) in urine or serum to aid in the early detectionofpregnancy.
Human chorionic gonadotropin (hCG) is a glycoprotein hormone produced by the developing placenta shortly after fertilization. In normal pregnancy, hCG can be detected in both urine and serum or plasma as early as 7 to 10 days after conception. hCG levels continue to rise very rapidly, frequently exceeding 100mIU/ml by the first missed menstrual period, and peaking in the 100,000-200,000mIU/ml range about 10-12 weeks into pregnancy. The appearance of hCG in both the urine and serum or plasma soon after conception, and its subsequent rapid rise in concentration during early gestational growth, make it an excellent marker for the early detection of pregnancy. The hCG Pregnancy Rapid Test Cassette is a rapid test that qualitatively detects the presence of hCG in urine or serum or plasma specimen at the sensitivity of 25mIU/ml. The test utilizes a combination of monoclonal and polyclonal antibodies to selectively detect elevated levels of hCG in urine or serum or plasma. At the level of claimed sensitivity, the hCG Pregnancy Rapid Test Cassette shows no cross-reactivity interference from the structurally related glycoprotein hormones hFSH, hLH and hTSH at high physiological levels.
1. Bring the pouch to room temperature (15-30℃) before opening it. Remove the cassette from the sealed pouch and use it within one hour.
2. Place the cassette on a clean and level surface. Hold the dropper vertically and transfer 3 full drops of urine or serum or plasma (approximately 120ul) to the specimen well of the cassette, and then start the timer. Avoid trapping air bubbles in the specimen well. See illustration below.
3. Wait for the colored line(s) to appear. Read the result at 3 minutes when testing a urine specimen, or at 5 minutes when testing a serum or plasma specimen.
INTERPREATION OF RESULT:
POSITIVE: Two distinct red lines appear. One line should be in the control region(C)andanotherlineshouldbeinthetestregion(T).
NEGATIVE: One red line appears in the control region (C). No apparent redorpinklineappearsinthetestregion.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test strip. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
|Acetylsalicylic Acid||20 mg/ml|
|Ascorbic Acid||20 mg/mL|
|Bilirubin (serum)||Bilirubin||40 mg/dL|
|Gentisic Acid||20 mg/mL|
|Triglycerides (serum)||1200 mg/dL|
|ORIENT NEW LIFE MEDICAL CO., LTD.|
|Email:||Jerry @ newlifebiotest .com|
Contact Person: Jerry Meng