Payment & Shipping Terms:
|Category:||HOME Fertility Test||Format:||Cassette Type|
|Specimen:||Urine||Testing Time:||5-15 Minutes|
|Shelf Life:||24 Months||Application:||Pregnancy Test Home|
reliable pregnancy test kit,
pregnancy ovulation kit
One step LH Ovulation Rapid test cassette, Gold Colloidal method urine specimen, easily use
The LH One Step Ovulation Test Device (Urine) is a rapid chromatographic immunoassay for the qualitative detection of luteinizing hormone (LH) in urine to aid in the detection of ovulation.
Ovulation is the release of an egg from the ovary. The egg then passes into the fallopian tube where it is ready to be fertilized. In order for pregnancy to occur, the egg must be fertilized by sperm within 24 hours after its release. Immediately prior to ovulation, the body produces a large amount of luteinizing hormone (LH) which triggers the release of a ripened egg from the ovary. This “LH surge” usually takes place in the middle of the menstrual cycle.
The LH One Step Ovulation Test Device (Urine) is a complete system to help predict the time of ovulation, and peak fertility. It is during this fertile time that pregnancy is most likely to occur.
The LH One Step Ovulation Test Device (Urine) detects the LH surge in urine, signaling that ovulation is likely to occur in the next 24-36 hours. The test utilizes a combination of antibodies including a monoclonal LH antibody to selectively detect elevated levels of LH.
Important: The LH surge and ovulation may not occur in all cycles.
INTERPRETATION OF RESULTS
POSITIVE: Two lines are visible, and the line in test line region (T) is the same as or darker than the one in the control line region (C). This indicates probable ovulation in 24-36 hours.
NEGATIVE: Two lines are visible, but the line in the test line region (T) is lighter than the one in the control line region (C), or if there is no line in the test line region (T). This indicates that no LH surge has been detected.
INVALID: Control line fails to appear. Insufficient specimen volume or incorrect procedural techniques are the most likely reasons for control line failure. Review the procedure and repeat the test with a new test. If the problem persists, discontinue using the test kit immediately and contact your local distributor.
TEST RESULTS CHART
|ORIENT NEW LIFE MEDICAL CO., LTD.|
|Email:||Jerry @ newlifebiotest .com|
Contact Person: Jerry Meng